Find great deals for Endress Hauser Prosonic FMU Ultrasonic Transmitter Measuring System. Shop with confidence on eBay!. matrix fmu – Download as PDF File .pdf), Text File .txt) or read online. FMU Ultrasonic Measurement Operating Instructions. FMU FMU FMU Endress + Hauser The Power of Know How.
|Published (Last):||28 August 2009|
|PDF File Size:||1.36 Mb|
|ePub File Size:||4.91 Mb|
|Price:||Free* [*Free Regsitration Required]|
Endress+Hauser Prosonic FMU 860 Manuals
Ciudad de Guatemala, C. The LED of the relay lights up when the relay energises. The sensors are supplied with a permanently attached cable up to 30 m, cable diameter0.
QV 40,09 1. EmptyThis module must be applied during configuration see belowif main value 2 is not to appear in the data telegram. Touch an earthed object before handling the board. This pulse is reflectedback from the surface of the product.
Endresse Hauser FMU 860 Ultrasonic measurement
When the mounting fmi are very unfavourable, it may be necessary tosuppress noise signals see chapter 9. After determining the length units, this can only be altered when all otherparameters are also changed. Controls Prosonic FMU fum Home Documents Manual Prosonic Fmu Signal below Signal above Allother modes are described in chapter 4. As long as the basic settings have not been concluded, the Prosonic FMU emits awarning message. The relay is in the de-energized state dropped out when the normally closedcontact is closed.
On FMUthe two current outputs are electrically connected,likewise the two sensors inputs. Mounting theall-weather protective cover on the IP 66 protective fmi is also illustrated. If a number cannot be displayed on the 4 12 digit display, E appears. With the first pair of values the smallest volume to be measured and thecorresponding level should be entered.
This shielding serves as a return cable and corresponds to the black wire of the unshortened cable. Reset the Prosonic FMU.
Close operating plate and tighten screws. With thick controlcabinets, the fixing points at therear may be used. Signal pattern recognition using fuzzy logic elements and selectable applicationparameters to shorten commissioning times and to ensure long-term and correctultrasonic measurement.
Vmu version is for differential measurement or collecting data from individualmeasuring points. Units of length may only be altered immediately after a reset of the transmitter. Ensure that the power is switched off when the unit is connected up. Devices located in safe areas still require a certificate if their outputs run into explosion hazardous areas.
Move to the next digit by pressing again. Depending on the height at which it is mounted, the limit switch acts as a minimum limitor a maximum limit switch. This appears in the device data base file name. In the case of the Prosonic FMU the external limit switch canbe assigned to either channel.
Check that the power supply for the instrument agrees with that on the nameplate. FMU 86 Order codeChapter 1: Loosen the metallic braiding, twist it together securely and connect it to terminal 80 Channel 1 or terminal 90 Channel 2 Fig. First time users are recommended to read thoroughly the operating instructions, whichapply to their application. After entering the point number the asigned point of values, level and flow, can bedisplayed The more pairs of values you enter, the more exact the linearization will be.
Additional bitmap files are required in order to represent the device by an icon in the network design software. Note the settings When entering the parameters the values entered can be noted in the table on page The counter can not be transmitted. Codes to are reserved for customer-specific weirs. The parameters are set usingsubmenus.
Endress+hauser Prosonic FMU Manuals
In a protective housing IP 66 for wall or post mounting in the field or control room In a separate operating unit for control panel mounting, when the board withthe transmitter electronics can be mounted separately.
Two important rules for linearization must be observed: Safe area non-explosion hazardous area Symbol used in drawings to indicate, if necessary, non-explosion hazardous areas. The following entries are tmu Application parameter 2Fine bulk solidsApplication parameter 2, see Fig. The beginning and end of the range can be definedarbitrarily, i.